Most Active Hardware Boards: Message 7 of 8. Thank you for your reply. I connected the device to the labview. It doesnt have the outputs. What I am trying to do now is: I used probes for debugging.
|Date Added:||18 June 2008|
|File Size:||70.75 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
A frequency sweep from F1 to Fn then obtain 2 different output such as C and Labivew for each frequency using Agilent E driver vi’s. Thank you for your consideration and time dear Dale. Do you know how I should modify this example? You should be able to find it here: Message 3 of 8.
Agilent EA LCR meter automation using Labview. – Discussion Forums – National Instruments
I used probes for debugging. Message 7 of 8. I can not find the reason. The array consists of numbers and text like:. Message 8 of 8.
Hello, I want to measure capacitance and resistance in a frequency domain. Message 4 of 8. I am trying to do a frequency sweep. It sweeps between 3 frequencies and has high and low limit bands which I dont know what these bands are.
It looks like the drivers can overcome the point limit for you from what I can tell. I am using keysight driver Labview package instead of Visa.
The array consists of numbers and text like: But i have problems with input and output of these vi’s.
Thank you in advance. After finalizing it I will publish it here for those who one day may need it.
Instrument Control (GPIB, Serial, VISA, IVI)
I want to measure capacitance and resistance in a frequency domain. Message 2 of 8. The example seems to gather data points between Hz and Hz, then data points between Hz and Hz and then data points between Hz and Hz. The maximum number of measued data agileny in this LCR meter.
Message 5 of 8. I would like to verify that this is correct, but I do not actually have the hardware available to me.
Agilent Technologies / Keysight Technologies E4980A
What I am trying to do now is: Most Active Hardware Boards: Inside the code I have a problem: You should be able to find it here:. I’ve had a chance to look at the example a bit more closely now. Have you had a look at the Read Measurement List Sweep example?
It does not give the data. I wrote a code which uses single mesurment command in a FOR loop which adds a specific amount to frequency amount and then again conducts another single mesurment and gave the results at the end of loop.
need help: Agilent EA LCR meter through GPI | Keysight Community
There are also links to more specific examples at the bottom of that page. Once you have the data collected and stored as an array perhaps, you can write to a spreadsheet with one of the VIs mentioned here:. Thank you for your reply.